lanxicy.com

第一范文网 文档专家

第一范文网 文档专家

ARTICLE IN PRESS

Optics and Lasers in Engineering 47 (2009) 282–291 www.elsevier.com/locate/optlaseng

3-D computer vision in experimental mechanics

Jean-Jose Orteu

Ecole Mines Albi, Campus Jarlard, 81013 Albi, France Available online 11 January 2008

Abstract Optical methods that give displacement or strain elds are now widely used in experimental mechanics. Some of the methods can only measure in-plane displacements/strains on planar specimens and some of them can give both in-plane and out-of-plane displacement/ strain elds on any kind of specimen (planar or not). In the present paper, the stereovision technique that uses two cameras to measure 3-D displacement/strain elds on any 3-D object is presented. Additionally, a quite inclusive list of references on applications of stereovision (and 3-D DIC) to experimental mechanics is given at the end of the paper. r 2007 Elsevier Ltd. All rights reserved.

Keywords: Stereovision; Stereo-correlation; 3-D digital image correlation (3D-DIC); Shape measurement; Displacement/strain measurement; Experimental mechanics

1. Introduction Full-eld optical techniques for displacement or strain measurements are now widely used in experimental mechanics. The main techniques are photoelasticity, geometric moire, moire interferometry, holographic interferometry, speckle interferometry (ESPI), the grid method and digital image correlation (DIC) [1–9]. It should be noted that some of these techniques can only measure inplane displacements/strains on planar specimens and some of them can give both in-plane and out-of-plane displacement/strain elds on any kind of specimen (planar or not). Due to its (apparent) simplicity and versatility, the DIC method is probably one of the most commonly used methods, and many applications can be found in the literature1 [10–45]. When it is used with a single camera (classical DIC), the DIC method can only give in-plane displacement/strain elds on planar objects. By using two cameras (stereovision), the 3-D displacement eld and the

Tel.: +33 563 493 073; fax: +33 563 493 099.

surface strain eld of any 3-D object can be measured. Using stereovision in conjunction with DIC leads to a socalled digital image stereo-correlation technique (DISC), also called 3-D DIC. In this paper, the stereovision (and 3-D DIC) technique developed at the Research Center on Tools, Materials and Forming Processes (CROMeP) at Ecole des Mines d’Albi is presented. 2. Stereovision for 3-D shape and 3-D displacement/strain measurement Binocular stereovision is a technique for recovering the 3-D structure of a scene from two different viewpoints (see Fig. 1 where PX ; Y ; Z is the 3-D point to be measured, p1 u1 ; v1 and p2 u2 ; v2 are its stereo projections in the images, C 1 and C 2 are the optical centers of the two cameras). From a pair of images, it is possible to compute the 3-D coordinates of a physical 3-D point by triangulation assuming that: 1. The geometry of the stereo rig (i.e. the relative position and orientation of the two cameras) is known. This problem is solved by an off-line camera calibration procedure.

E-mail address: Jean-Jose.Orteu@enstimac.fr URL: http://www.enstimac.fr/ 1 To date, the author has noted more than 350 journal papers dealing with DIC-based measurements in experimental mechanics. The reference list can be provided upon request. 0143-8166/$ - see front matter r 2007 Elsevier Ltd. All rights reserved. doi:10.1016/j.optlaseng.2007.11.009

ARTICLE IN PRESS

J.-J. Orteu / Optics and Lasers in Engineering 47 (2009) 282–291 283

C2

C1 p2 (u2,v2) p1 (u1,v1)

P (X,Y,Z)

Fig. 2. The stereo rig used in our lab.

Fig. 1. Binocular stereovision.

2. The two image points p1 and p2 are matched, i.e. identied as corresponding to the same physical point P. This is called the stereo-matching problem. 2.1. Calibration of the stereovision sensor Camera calibration is an important task in 3-D computer vision, particularly when metric data are required for applications involving accurate dimensional measurements. Calibrating a camera involves determining its intrinsic parameters (generally, the two coordinates in the image frame of the intersection of the optical axis with the image plane, the two scale factors along the vertical and horizontal axes of the image frame, and, if need be, the lens distortion parameters) and its position and orientation with respect to an arbitrary world reference frame [46]. Calibrating a stereovision sensor made up of two cameras involves determining the intrinsic parameters of each camera and the relative position and orientation between the two cameras (see Fig. 2). These calibration data are required to compute, by triangulation, the 3-D coordinates of a point corresponding to matched pixels on the two images. An original and exible technique has been developed to easily calibrate a camera or a stereovision sensor. Lens distortion is taken into account. The technique only requires the camera to observe a (planar) pattern shown at a few different orientations (typically between 10 and 20). The motion of the pattern need not be known and the pattern itself can be imprecise (i.e. its geometry need not be known accurately). Using a photogrammetric bundle adjustment approach [47], the intrinsic parameters of each camera (including the distortion parameters), the 3-D points of the pattern and the relative position and

orientation between the two cameras are estimated all together. The technique is easy to use, exible and leads to a highly accurate calibration [48,49]. The calibration parameters will be used throughout this paper at different stages for:

the rectication of the pairs of stereo images; the correction of lens distortion; the calculation of the 3-D position of a scene point from its stereo projections by triangulation.

2.2. Rectication of the pair of stereo images Given a point p1 in image 1, its corresponding point p2 in image 2, called stereo-correspondant (this is the stereomatching problem discussed in Section 2.3) appears to be always lying along a line of image 2 entirely dened by the coordinates of p1 , the relative position/orientation of the two cameras and their intrinsic parameters: this is the socalled epipolar line associated to p1 [50,51] (see Fig. 3). This geometric constraint inherent to any stereo imaging system, called the epipolar constraint, is very interesting since it simplies the search for the stereo-correspondant of a given point from a 2-D search across the entire image to a 1-D search along its epipolar line. In the particular case where the cameras are perfectly aligned, the epipolar lines are parallel to the image rows (see Fig. 4): this ideal conguration simplies the search for correspondence since corresponding pixels are on the same row in both images. In this conguration, the difference between the column coordinates of two matched points is called ‘‘disparity’’. In practice, the mechanical alignment of the cameras is very difcult to obtain. Nevertheless, the problem can be solved at the processing level by applying a transformation (plane-to-plane homography2) to each image of the initial

2 A plane-to-plane homography is a bijective projective transformation mapping pairs of points between planes [51,52].

ARTICLE IN PRESS

284 J.-J. Orteu / Optics and Lasers in Engineering 47 (2009) 282–291

numerous algorithms for image matching have been proposed. They can be classied into two categories: 1. Feature matching: The algorithms rst extract salient primitives from the images, such as edge segments or contours, and match them in the views being considered. These methods are fast because only a small subset of the image pixels is used but usually give few matches (sparse disparity maps). 2. Template matching: The algorithms attempt to correlate the grey levels of image patches in the views being considered, assuming that they present some similarity. The underlying assumption appears to be a valid one for relatively textured areas and for image pairs with small differences. These methods can give dense disparity maps. In [54], a method based on stereovision for measuring strains in stamped 3-D sheet metal parts was proposed. This method falls into category 1 as it requires that a predened pattern (a grid of squares) be applied to the sheet surface before stamping. Each image of the stereo pair is processed independently in order to locate the grid intersections. The grid intersections are extracted in such a way as to allow an automatic matching between the two images, provided that the operator matches manually a single pair of points. A stereo-correlation technique that falls into category 2 has also been developed (and is presented below). With this technique, a regular grid need not be applied to the object and the meshes used to compute the local strains are generated at the post-processing level (meshes of any size can be generated, and in particular small ones to take into account the large strain gradients). Correlation scores are computed by measuring the similarity of a xed window in the rst image to a shifting window in the second (see Fig. 5). The second window is moved into the second image by integer increments along the corresponding image row (remember that a ‘‘rectied space’’ is being worked with) and a curve of correlation scores is generated (see Fig. 6). The correct matching corresponds to the highest peak provided that this peak is greater than a threshold (S min ). Note that a match is not accepted if the highest and the second highest peaks are

Left Image Right Image

Fig. 3. The camera optical centers C 1 and C 2 , the 3-D point P and its images p1 and p2 lie in a common plane. This plane intersects each image plane in an epipolar line.

Fig. 4. A rectied stereovision sensor: the epipolar lines are parallel to the image rows.

stereo pair to obtain a new pair of stereo images corresponding to a virtual stereo rig with perfectly aligned cameras [50,51]. This rectication procedure uses the calibration parameters computed in the off-line camera calibration phase. Notice that during the rectication, the real distorted images (due to optical lens distortion) are also transformed into ideal distortion-free images. The image rectication procedure results in a pair of images that corresponds to an ideal stereo rig of distortion free and perfectly aligned cameras. This greatly simplies the stereo-matching phase. It should be noted that, in practice, neither the rectied images nor the undistorted images are computed. In fact, the computed rectication homographies and the distortion parameters are used directly in the expression of the correlation function to work on the raw images. This new approach can achieve the highest matching accuracy by avoiding the pixel interpolations involved in the distortion correction and rectication of the images (see [49,53] for more details). 2.3. Stereo-matching The main difculty in stereovision is to establish correspondences between pairs of images. Over the years,

Fig. 5. Correlation-based stereo-matching.

ARTICLE IN PRESS

J.-J. Orteu / Optics and Lasers in Engineering 47 (2009) 282–291 285

1.0 correlation score value 0.8 0.6 0.4 0.2 0.0 0.2 0.4 0.6 5

smin

Δ

[55] can be used to compute the 3-D position of a scene point corresponding to a stereo pair of image points (see Fig. 1), assuming that their distortion have been corrected. By repeating this operation for a large number of stereo matched pairs the 3-D shape of an object can be obtained. 2.5. 3-D displacement measurement

10

15 20 25 disparity value [pixel]

30

35

Fig. 6. Curve of correlation scores.

Fig. 7. Projective deformation: the segment of length L in space is projected to a segment of length l 1 in the left image and to a segment of length l 2 al 1 in the right image.

within a minimum range (D). By the denition, corresponding points have coordinates u; v and u; v d in left and right rectied images, and d, the distance in pixels, is the disparity (see Fig. 5). Fig. 5 presents a gross simplication of stereo-matching process: matching windows are assumed to be square and of the same size. In practice, due to local surface non-planarity as well as projective distortion, a square window is most likely matched with a non-square one as shown in Fig. 7. So, the correlation between the xed left window (a rectangle) and the shifting right window (a distorted rectangle) is calculated. The optimal shape of the correlation window in the right image is computed according to the local orientation and curvature of the surface. In Fig. 6, the disparity is an integer value but in reality a subpixel correlation-based stereomatching algorithm has been developed so the disparity is a real value. For more details see [49,53]. As mentioned above, it is well known that the correlation technique is efcient on textured objects. When an object does not present a suitable random, contrasted texture, one may apply a thin coating of high-contrast particles such as from spray painting [11], toner powder, ink, lithography, etc. 2.4. 3-D reconstruction Using the calibration parameters of each camera and the rectifying homographies, a classical triangulation method

Using the stereovision technique, the shape variation of an object can be measured by analyzing a sequence of pair of stereo images. Nevertheless, in experimental mechanics, we are generally interested in the surface strain eld, that can be obtained by tracking the displacement of some points at the surface of an object undergoing some mechanical or thermal stress. As already mentioned, much work has been done on 2-D displacement/strain measurement, using a single camera, on gridded objects or using correlation-based techniques. These methods can give only the in-plane strains. The stereovision technique allows measurement of the 3-D displacement eld of an object undergoing 3-D deformation. From the pair of stereo images taken at time t0 , the 3-D shape of the object at time t0 can be computed. From the pair of stereo images taken at time t1 , the 3-D shape of the object at time t1 can be computed. In addition, by matching the two images taken by the left camera (or the right one) at time t0 and t1 (this is called temporal matching or tracking), the 3-D displacement corresponding to each image point can be computed (see Fig. 8). The temporal matching problem is similar to the classical 2-D displacement measurement problem. It is important to note that, in this case, the epipolar constraint cannot be used since two time-varying images taken by a single camera are being dealt with. Depending on whether the object is marked with a grid or not the temporal matching problem can be tackled using feature matching techniques or a correlation-based algorithm. When 3-D displacement elds are measured by 3-D DIC, stereo-correlation (i.e. stereovision + correlation-based stereo-matching), that gives 3-D information, and pixel tracking by correlation (DIC) in the sequence of stereo images are used in conjunction. In order to achieve better accuracy an improved method (combining temporal tracking and

Fig. 8. 3-D displacement eld computation.

ARTICLE IN PRESS

286 J.-J. Orteu / Optics and Lasers in Engineering 47 (2009) 282–291

stereo-correlation) has been proposed that operates simultaneously on the two pairs of stereo images [53]. 2.6. 3-D strain computation In [54], the strain eld was computed by analyzing the initial and nal conguration of an array of points marked on the surface. Each square element of the grid pattern was divided into triangular sub-elements. The principal strains in each element were determined by comparing the undeformed and deformed triangles. With 3-D DIC, no regular grid is marked on the surface of the object but the 3-D mesh resulting from the 3-D reconstruction of a Delaunay triangular mesh dened in an image can be computed (sparse reconstruction, see Fig. 9). The strain eld can be computed by using each 3-D triangular element of the meshes before and after the deformation. A niteelement software can be used for these computations. In our laboratory, the software Abaqust is often used for these computations. A 3-D triangular mesh is generated from the 3-D points and Abaqust computes the strain values from the 3-D displacements imposed at the nodes of the mesh [56]. 2.7. Accuracy The accuracy of 3-D reconstruction depends on many factors:

divided by the size of the object) achieved using stereocorrelation under controlled laboratory conditions is about 1/50 000 (see [49] for a more detailed discussion on 3-D reconstruction accuracy). The strains are computed by numerical derivation from the displacements measured at several points distributed over a mesh. The strain at each point of the mesh is computed using neighboring points (see Fig. 9). The accuracy of the strain computation depends on many factors: the discrete derivation scheme used (shape of the integration domain) [57], the accuracy of the displacement measurement, and the computation basis (mesh element spacing). The method proposed can reach an accuracy of 0:05% 500 mstrain on the computed strains with triangular elements measuring 50 pixels. For example, using the integration domain shown in Fig. 9 (four neighboring points) and the discrete derivation scheme described in [57], the uncertainty for a 2-D strain computation can be evaluated using the following relation: Dd D p , 2 Dx where D is the uncertainty on strain computation, Dd the uncertainty on displacement measurement and Dx the computation basis for the discrete derivation. For example, with Dd equal to 1/100 pixel and a derivation basis Dx equal to 14 pixels, the uncertainty on strain measurement D is equal to 0.05%. An uncertainty 10 times lower (0.005%) can be achieved by taking a derivation basis 10 times larger (140 pixels) but the result has much less local signicance, which can be a problem with heterogeneous materials. 3. Conclusion Stereovision (and 3-D DIC) has proven to be a powerful non-contact technique for measuring 3-D displacement/ strain elds on any 3-D object and is now widely used for industrial applications. Several commercial software are available [163–166]. The technique is very versatile and can be used for a large scale of experimental mechanics problems: it can be applied at micro or macro scale (small or large structures), it can measure both small (a few hundred of microstrains) and large strains 4200%, it can be used for high-speed dynamic tests by using high-speed cameras, etc. During the talk,3 several examples of application will be presented (sheet metal forming, polymer forming, refractory concrete or composite structures mechanical behavior characterization, bio-mechanics, high-speed impact tests, etc.). Additionally, a quite inclusive list of references on applications of stereovision (and 3-D DIC) to experimental mechanics is given at the end of this paper [58–162].

This paper is the written support of an Invited Keynote Lecture given at the 3rd Workshop on Optical Measurement Techniques for Structures and Systems, Leuven, Belgium, 28–30 May 2007.

3

the quality of the cameras and their resolution; the conguration of the two cameras (distance between the two cameras, angle between their optical axis, etc.) which governs the triangulation accuracy; the accuracy of the stereovision sensor calibration; the accuracy of the matched features in the images.

The accuracy of the matched features depends on the type of extracted feature. In [54] gridded objects were used and the grid intersection extraction accuracy was 1/30 pixel. In [49] it has been shown that the DIC technique leads to an accuracy better than 1/100 pixel. Using high resolution digital cameras with 1024 1024 pixels each, the 3-D reconstruction relative accuracy in object space (the accuracy of coordinate determination

Fig. 9. 3-D strain eld computation.

ARTICLE IN PRESS

J.-J. Orteu / Optics and Lasers in Engineering 47 (2009) 282–291 287

References

[1] Cloud G. Optical methods of engineering analysis. New York: Cambridge University Press; 1998. [2] Rastogi P, editor. Photomechanics. Berlin: Springer; 1999. ISBN 3-540-65990-0. [3] Feldmann O, Mayinger F. Optical measurements. In: Heat and mass transfer, second ed. Berlin: Springer; 2001. ISBN 978-3-540-66690-5. [4] Molimard J, Lee JR, Surrel Y, Vautrin A. Full-eld optical techniques: applications to strain measurement and mechanical identication. In: Invited keynote lecture at the 10th European conference on composite materials (ECCM10), Bruges, Belgium, 2002. [5] Surrel Y. Full-eld optical methods for mechanical engineering: essential concepts to nd one’s way. In: Invited keynote lecture at the second international conference on composites testing and model identication, Bristol, UK, 2004. [6] Grediac M. The use of full-eld measurement methods in composite material characterization: interest and limitations. Composites Part A 2004;35:751–61. [7] Dally JW, Riley WF. Experimental stress analysis, fourth ed. College House Enterprises, LLC, Knoxville, USA, 2005. ISBN 0-9762413-0-7. [8] Doyle J. Modern experimental stress analysis. New York: Wiley; 2004 ISBN-10: 0-470-86156-8. [9] Sharpe J, William N, editors. Springer handbook of experimental solid mechanics. Berlin: Springer; 2007. ISBN 0-387-30877-6. [10] Peters WH, Ranson WF. Digital imaging techniques in experimental stress analysis. Opt Eng 1982;21(3):427–31. [11] Sutton MA, Wolters WJ, Peters WH, Ranson WF, McNeill SR. Determination of displacements using an improved digital correlation method. Image Vision Comput 1983;1(3):133–9. [12] Chu TC, Ranson WF, Sutton MA, Peters WH. Applications of digital-image-correlation techniques to experimental mechanics. Exp Mech 1985;25(3):232–44. [13] McNeill SR, Peters WH, Sutton MA. Estimation of stress intensity factor by digital image correlation. Eng Fract Mech 1987;28(1): 101–12. [14] Chao YJ, Sutton MA. Measurement of strains in a paper tensile specimen using computer vision and digital image correlation—part 1: data acquisition and image analysis system. Tappi J 1988;70(3): 173–5. [15] Kahn-Jetter ZL, Jha NK, Bhatia H. Optimal image correlation in experimental mechanics. Opt Eng 1994;33(4):1099–105. [16] Han G, Sutton MA, Chao YJ. A study of stationary crack-tip deformation elds in thin sheets by computer vision. Exp Mech 1994;34(2):125–40. [17] Zink AG, Davidson RW, Hanna RB. Strain measurement in wood using a digital image correlation technique. Wood Fiber Sci 1995; 27(4):346–59. [18] Lyons JS, Liu J, Sutton MA. High-temperature deformation measurements using digital-image correlation. Exp Mech 1996; 36(1):64–70. [19] Sun Z, Lyons JS, McNeill SR. Measuring microscopic deformations with digital image correlation. Opt Lasers Eng 1997;27(4): 409–28. [20] Choi S, Shah SP. Measurement of deformations on concrete subjected to compression using image correlation. Exp Mech 1997;37(3):307–13. [21] Vendroux G, Knauss WG. Submicron deformation eld: part 2. Improved digital image correlation. Exp Mech 1998;38(2):86–92. [22] Sutton MA, McNeill SR, Helm JD, Chao YJ. Advances in twodimensional and three-dimensional computer vision. In: Rastogi PK, editor. Photomechanics, topics in applied physics. Berlin: Springer; 2000. [23] Wattrisse B, Chrysochoos A, Muracciole J-M, Nemoz-Gaillard M. Analysis of strain localization during tensile tests by digital image correlation. Exp Mech 2001;41(1):29–39.

[24] Chevalier L, Calloch S, Hild F, Marco Y. Digital image correlation used to analyze the multiaxial behavior of rubber-like materials. Eur J Mech—A/Solids 2001;20:169–87. [25] Wang Y, Cuitino AM. Full-eld measurements of heterogeneous deformation patterns on polymeric foams using digital image correlation. Int J Solids Struct 2002;39(13):3777–96. [26] Abanto-Bueno J, Lambros J. Investigation of crack growth in functionally graded materials using digital image correlation. Eng Frac Mech 2002;69(14):1695–711. [27] Zhang D, Eggleton CD, Arola DD. Evaluating the mechanical behavior of arterial tissue using digital image correlation. Exp Mech 2002;42(4):409–16. [28] Knauss WG, Chasiotis I, Huang Y. Mechanical measurements at the micron and nanometer scales. Mech Mater 2003;35(3):217–31. [29] Barthelat F, Wu Z, Prorok BC, Espinosa HD. Dynamic torsion testing of nanocrystalline coatings using high-speed photography and digital image correlation. Exp Mech 2003;43(3):331–40. [30] Li EB, Tieu AK, Yuen WYD. Application of digital image correlation technique to dynamic measurement of the velocity eld in the deformation zone in cold rolling. Opt Lasers Eng 2003; 39(4):479–88. [31] Zhang J, Xiong C, Li H, Li M, Wang J, Fang J. Damage and fracture evaluation of granular composite materials by digital image correlation method. Acta Mech Sin 2004;20(4):408–17. [32] Parsons E, Boyce MC, Parks DM. An experimental investigation of the large-strain tensile behavior of neat and rubber-toughened polycarbonate. Polymer 2004;45:2665–84. [33] Bergonnier S, Hild F, Roux S. Digital image correlation used for mechanical tests on crimped glass wool samples. J Strain Anal Eng Des 2005;40(2):185–97. [34] Ambu R, Aymerich F, Bertolino F. Investigation of the effect of damage on the strength of notched composite laminates by digital image correlation. J Strain Anal Eng Des 2005;40(5):451–62. [35] Chang S, Wang CS, Xiong CY, Fang J. Nanoscale in-plane displacement evaluation by AFM scanning and digital image correlation processing. Nanotechnology 2005;16(4):344–9. [36] Jin H, Bruck HA. A new method for characterizing nonlinearity in scanning probe microscopes using digital image correlation. Nanotechnology 2005;16(9):1849–55. [37] Yoneyama S, Morimoto Y, Takashi M. Automatic evaluation of mixed-mode stress intensity factors utilizing digital image correlation. Strain 2006;42(1):21–9. [38] Hild F, Roux S. Digital image correlation: from displacement measurement to identication of elastic properties—a review. Strain 2006;42(2):69–80. [39] Ait-Amokhtar H, Vacher P, Boudrahem S. Kinematics elds and spatial activity of Portevin-Le Chatelier bands using the digital image correlation method. Acta Mater 2006;54(16):4365–71. [40] Boyce BL, Reu PL, Robino CV. The constitutive behavior of laser welds in 304L stainless steel determined by digital image correlation. Metall Mater Trans A 2006;37(8):2481–92. [41] Yaofeng S, Pang JHL. AFM image reconstruction for deformation measurements by digital image correlation. Nanotechnology 2006;17(4):933–9. [42] Bereld TA, Patel JK, Shimmin RG, Braun PV, Lambros J, Sottos NR. Micro- and nanoscale deformation measurement of surface and internal planes via digital image correlation. Exp Mech 2007;47(1): 51–62. [43] Fayolle X, Calloch S, Hild F. Controlling testing machines with digital image correlation. Exp Tech 2007;31(3):57–63. [44] Li X, Xu W, Sutton MA, Mello M. In situ nanoscale in-plane deformation studies of ultrathin polymeric lms during tensile deformation using atomic force microscopy and digital image correlation techniques. IEEE Trans Nanotechnology 2007;6(1): 4–12. [45] Yoneyama S, Kitagawa A, Iwata S, Tani K, Kikuta H. Bridge deection measurement using digital image correlation. Exp Tech 2007;31(1):34–40.

ARTICLE IN PRESS

288 J.-J. Orteu / Optics and Lasers in Engineering 47 (2009) 282–291 [65] Boochs F. Off-line compilation of photogrammetric stereo models using digital image correlation. Photogrammetria 1987;41(3): 183–99. [66] Bruck HA, Schreier HW, Sutton MA, Chao YJ, Deng X, Davoud M. Distortion in GMAW of thin plates: temperature and 3-D deformation measurements using high-speed thermal imaging and stereoscopic video imaging. In: Proceedings of fth international conference on trends in welding research, Pine Mountain, GA, USA, 1998, p. 967–71. [67] Burguete R. Keynote presentation: industrial applications of high speed 3-D DIC to structural tests. In: Proceedings of SEM 2007 annual conference on experimental and applied mechanics, Springeld, MA, USA, 2007. [68] Cardenas-Garcia JF, Yao HG, Zheng S. 3D reconstruction of objects using stereo imaging. Opt Lasers in Eng 1995;22(3):193–213. [69] Choi S, Shah SP. Measurement of deformations on concrete subjected to compression using image correlation. Exp Mech 1997;37(3):307–13. [70] Compston P, Styles M, Kalyanasundaram S. Low energy impact damage modes in aluminum foam and polymer foam sandwich structures. J Sandwich Struct Mater 2006;8(5):365–79. [71] Cornille N, Garcia D, Sutton MA, McNeill SR, Orteu J-J. Automated 3-D reconstruction using a scanning electron microscope. In: Proceedings of SEM 2003 annual conference on experimental and applied mechanics, Charlotte, North Carolina, USA, 2003. [72] Cornille N, Garcia D, Sutton MA, McNeill SR, Orteu J-J. 3D shape and strain measurements using a scanning electron microscope. In: Proceedings of international conference on experimental mechanics (ICEM12), Bari, Italy, 2004. [73] Cornille N. Accurate 3D shape and displacement measurement using a scanning electron microscope. PhD thesis, INSA, France and University of South Carolina, Columbia, USA; June 2005. [74] Davies PA, Randle V. Combined application of electron backscatter diffraction and stereo-photogrammetry in fractography studies. J Microsc 2001;204(1):29–38. [75] Florando JN, Rhee M, Arsenlis A, LeBlanc MM, Lassila DH. Calculation of the slip system activity in deformed zinc single crystals using digital 3-D image correlation data. Philos Mag Lett 2006;86(12):795–805. [76] Fournier N. Image stereo-correlation for the fast analysis of damages on aircrafts. In: Proceedings of SEM 2007 annual conference on experimental and applied mechanics, Springeld, MA, USA, 2007. [77] Galanulis K, Hofmann A. Determination of forming limit diagrams using an optical measurement system. In: Proceedings of seventh international conference on sheet metal—SheMet’99, Erlangen, Germany, 1999. [78] Garcia D, Orteu J-J. 3D deformation measurement using stereocorrelation applied to the forming of metal or elastomer sheets. In: Proceedings of international workshop on video-controlled materials testing and in situ microstructural characterization, Nancy, France, 1999. [79] Garcia D, Orteu J-J. 3D Deformation measurement using stereocorrelation applied to experimental mechanics. In: Proceedings of 10th FIG international symposium on deformation measurements, Orange, CA, USA, 2001. [80] Garcia D. Mesure de formes et de champs de deplacements tridimensionnels par stereo-correlation d’images. PhD thesis, Institut National Polytechnique de Toulouse, France, December 2001. [81] Garcia D, Orteu J-J, Cutard T, Cailleux E. Application of stereovision to the mechanical characterization of refractory ceramics reinforced with metallic bers. In: Proceedings of SEM 2002 annual conference on experimental and applied mechanics, Milwaukee, WI, USA, 2002. [82] Garcia D, Orteu J-J, Penazzi L. A combined temporal tracking and stereo-correlation technique for accurate measurement of 3D [46] Salvi J, Armangue X, Battle J. A comparative review of camera calibrating methods with accuracy evaluation. Pattern Recognition 2002;35:1617–35. [47] Triggs B, McLauchlan P, Hartley R, Fitzgibbon A. Bundle adjustment—a modern synthesis. In: Triggs B, Zisserman A, Szeliski R, editors. Vision algorithms. Berlin: Springer; 2000. [48] Garcia D, Orteu J-J, Devy M. Accurate calibration of a stereovision sensor: comparison of different approaches. In: Proceedings of vision, modeling, and visualization 2000, Saarbrucken, Germany, ¨ 2000. [49] Garcia D. Mesure de formes et de champs de deplacements tridimensionnels par stereo-correlation d’images, PhD thesis, Institut National Polytechnique de Toulouse, France; December 2001. [50] Ayache N, Hansen C. Rectication of images for binocular and trinocular stereovision. In: Proceedings of ninth IEEE international conference on pattern recognition, Roma, Italy, 1988, p. 11–6. [51] Loop C, Zhang Z. Computing rectifying homographies for stereo vision. Technical Report, Microsoft Research MSR-TR-99-21; 1999. [52] Faugeras OD. Three-dimensional computer vision: a geometric viewpoint. Cambridge, MA: The MIT Press; 1993. [53] Garcia D, Orteu J-J, Penazzi L. A combined temporal tracking and stereo-correlation technique for accurate measurement of 3D displacements: application to sheet metal forming. J Mat Process Technol 2002;125–126:736–42. [54] Orteu J-J, Garric V, Devy M. Camera calibration for 3D reconstruction: application to the measure of 3D deformations on sheet metal parts. In: SPIE proceedings of European symposium on lasers, optics and vision in manufacturing (EUROPTO), vol. 3101, Munich, Germany, 1997. [55] Hartley R, Sturm P. Triangulation. Comput Vision Image Understanding 1997;68(2):146–57. [56] Cornille N. Accurate 3D shape and displacement measurement using a scanning electron microscope. PhD thesis, INSA, France and University of South Carolina, Columbia, USA; June 2005. [57] Allais L, Bornert M, Bretheau T, Caldemaison D. Experimental characterization of the local strain eld in a heterogeneous elastoplastic material. Acta Metall Mater 1994;42(11):3865–80.

References on applications of stereovision and 3-D DIC to experimental mechanics

[58] Ackermann F. Digital image correlation: performance and potential application in photogrammetry. Photogramm Rec 1984;11(64): 429–39. [59] Albertani R, Stanford B. Hubner JP, Ifju PG. Aerodynamic coefcients and deformation measurements on exible micro air vehicle wings. Exp Mech 2007;47(5):625–35. [60] Alpers B, Bergmann D, Galanulis K, Winter D. Advanced deformation measurement in sheet metal forming. In: Proceedings of sixth international conference on technology of plasticity, Nuremberg, Germany, 1999. [61] Andresen K. Strain tensor for large three-dimensional surface deformation of sheet metal from an object grating. Exp Mech 1999;39(1):30–5. [62] Becker T, Splitthof K, Siebert T, Kletting P. Error estimations of digital image correlation measurements. In: Slangen P, Cerruti C, editors. SPIE proceedings of speckle06 conference, vol. 6341, N mes, France, 2006. [63] Berthaud Y, Torrenti JM, Fond C. Analysis of localization in brittle materials through optical techniques. Exp Mech 1997;37(2):216–20. [64] Berube KA, Lopez-Anido RA, Caccese V. Variability in exural response of E-Glass/vinyl ester composites fabricated using the VARTM process. In: Proceedings of SAMPE 2006, Long Beach, CA, USA, 2006.

ARTICLE IN PRESS

J.-J. Orteu / Optics and Lasers in Engineering 47 (2009) 282–291 displacements: application to sheet metal forming. J Mater Process Technol 2002;125–126:736–42. He X, Jiang M. 3D information acquired by the correlation of projected fringe patterns. In: Quan C, Chau FS, Asundi A, Wong BS, Lim CT, editors. SPIE proceedings of third international conference on experimental mechanics and third conference of the Asian committee on experimental mechanics, vol. 5852. Singapore, 2005. p. 257–63. Helava UV. Digital correlation in photogrammetric instruments. Photogrammetria 1978;34(1):18–41. Helm JD, McNeill SR, Sutton MA. Improved three-dimensional image correlation for surface displacement measurement. Opt Eng 1996;35(7):1911–20. Helm JD, Sutton MA, Boone ML. Characterizing crack growth in thin aluminum panels under tension–torsion loading using threedimensional digital image correlation. In: Lucas GF, McKeighan PC, Ransom JS, editors. Nontraditional methods of sensing stress, strain, and damage in materials and structures: second volume, ASTM STP 1323. American Society for Testing and Materials; 2001. Helm JD, Sutton MA, McNeill SR. Deformations in wide, centernotched, thin panels, part I: three-dimensional shape and deformation measurements by computer vision. Opt Eng 2003;42(5): 1293–305. Helm JD, Sutton MA, McNeill SR. Deformations in wide, centernotched, thin panels, part II: nite element analysis and comparison to experimental measurements. Opt Eng 2003;42(5):1306–20. Hemmleb H, Schubert M. Digital microphotogrammetry—determination of the topography of microstructures by scanning electron microscope. In: Proceedings of second Turkish–German joint geodetic days, Berlin, Germany, 1997. p. 745–52. Herbst C, Splitthof K, Ettemeyer A. New features in digital image correlation techniques. In: Proceedings of society of automotive engineers (SAE) world congress—SAE technical paper in SP-1957, Detroit, MI, USA, 2005. Hijazi A, Yardi N, Madhavan V. Determination of forming limit curves using 3D digital image correlation and in situ observation. In: Proceedings of SAMPE 2004, Long Beach, CA, USA, 2004. Holstein D, Salbut L, Kujawinska M, Juptner W. Hybrid ¨ experimental-numerical concept of residual stress analysis in laser weldments. Exp Mech 2001;41(4):343–50. Huang YH, Quan C, Tay CJ, Chen LJ. Shape measurement by the use of digital image correlation. Opt Eng 2005;44(8):087011. Jacobsson L, Enqvist B. Deformation measurement on rock specimen during Brazilian test using white light speckle photography. Technical Report 2004:38, SP Swedish National Testing and Research Institute; 2004. Janova D, Jan J. Robust surface reconstruction from stereo SEM images. In: Hlavac V, Sara R, editors. Computer analysis of images and patterns. Berlin: Springer; 1995. Jones A, Shaw J, WA. An experimental facility to measure the chemorheological response of inated elastomer membranes at high temperature. Exp Mech 2006;46(5):579–87. Kahn-Jetter ZL, Chu TC. Three-dimensional displacement measurements using digital image correlation and photogrammic analysis. Exp Mech 1990;30(1):10–6. Kim D-H, Lee J-H, Kang K-J. An experimental method to measure Q-parameter for two parameter theory of ductile fracture. In: Proceedings of ASME international mechanical engineering congress, Washington, DC, USA, 2003. Kolednik O, Stuwe HP. The stereophotogrammetric determination ¨ of the critical crack tip opening displacement. Eng Fract Mech 1985;21(1):145–55. Lacey AJ, Thacker NA, Crossley S, Yates RB. A multi-stage approach to the dense estimation of disparity from stereo SEM images. Image Vision Comput 1998;16(1):373–83. Langerak N, Atzema E, Essing J. Strain measurements with the PHAST system. In: International Deep Drawing Research Group 289 (IDDRG’2000)—meeting of the working groups, Ann Arbor, MI, USA, 2000. Larsson L, Sjodahl M, Thuvander F. Microscopic 3-D displacement ¨ eld measurements using digital speckle photography. Opt Lasers Eng 2004;41(5):767–77. Lawler JS, Keane DT, Shah SP. Measuring three-dimensional damage in concrete under compression. ACI Mater J 2001;98: 465–75. Lecompte D, Vantomme J, Sol H. Crack detection in a concrete beam using two different camera techniques. Struct Health Monit 2006;5(1):59–68. Lee J-H, Kang K-J. In situ measurement of lateral side-necking of a fracture specimen using a stereo vision and digital image correlation. J Korean Soc Precis Eng 2004;21(10):154–61. Lockwood WD, Reynolds AP. Use and verication of digital image correlation for automated 3-D surface characterization in the scanning electron microscope. Mater Charact 1999;42(2): 123–34. Lopez-Anido R, El-Chiti FW, Muszynski L, Dagher HJ, Thompson LD, Hess PE. Composite material testing using a 3-D digital image correlation system. In: Proceedings of composites 2004 conference, Tampa, FL, USA, 2004. Luo P-F. Measurement of three-dimensional crack tip deformation by digital image processing. PhD thesis, South Carolina University, Columbia, USA; 1992. Luo PF, Chao YJ, Sutton MA, Peters WH. Accurate measurement of three-dimensional deformations in deformable and rigid bodies using computer vision. Exp Mech 1993;33(2):123–32. Luo PF, Chao YJ, Sutton MA. Application of stereo vision to threedimensional deformation analyses in fracture experiments. Opt Eng 1994;33(3):981–90. Luo PF, Liou SS. Measurement of curved surface by stereo vision and error analysis. Opt Eng 1998;30(6):471–86. Luo PF, Huang FC. Application of stereo vision to the study of mixed-mode crack-tip deformations. Opt Lasers Eng 2000;33(5): 349–68. Luo PF, Chen JN. Measurement of curved-surface deformation in cylindrical coordinates. Exp Mech 2000;40(4):345–50. McGinnis MJ, Pessiki S, Turker H. Application of three-dimensional digital image correlation to the core-drilling method. Exp Mech 2005;45(4):359–67. McGinnis MJ, Pessiki S. Inuence of steel reinforcement on in-situ stress evaluation in concrete structures by the core-drilling method. In: Proceedings of the quantitative nondestructive evaluation conference, vol. 820. American Institut of Physics (AIP); 2006. p. 1358–65. McGowan DM, Ambur DR, Hanna TG, McNeill SR. Evaluating the compressive response of notched composite panels using fulleld displacements. J Aircr 2001;38(1):122–9. Medina-Cetina Z, Rechenmacher A, Image-based sensing of 3-D displacements for enhanced soil model calibration. In: DeGroot DJ, DeJong JT, Frost JD, Baise LG, editors. Proceedings of GeoCongress 2006, Atlanta, GA, USA, 2006. Miller TJ, Schreier HW, Reu PL. High-speed DIC data analysis from a shaking camera system. In: Proceedings of SEM 2007 annual conference on experimental and applied mechanics, Springeld, MA, USA, 2007. Montemayor E, Ray RP. Displacement measurements for cyclic triaxial tests using digital image correlation. In: Proceedings of GeoFrontiers 2005 congress, Austin, TX, USA, 2005. Morimoto Y, Fujigaki M. Automated analysis of 3-D shape and surface strain distributions of a moving object using stereo vision. Opt Lasers Eng 1993;18(3):195–212. Moser R, Lightner JG. Using three-dimensional digital imaging correlation techniques to validate tire nite-element model. Exp Tech 2007;31(4):29–36. Nam TH, Marvalova B. Three-dimensional displacement measurement of inated air-spring shell using stereo-correlation technique.

[83]

[102]

[103]

[84] [85]

[104]

[105]

[86]

[106]

[107]

[87]

[108]

[88]

[109]

[89]

[110]

[90]

[111] [112]

[91]

[113] [114]

[92]

[93] [94]

[115]

[116]

[95]

[117]

[96]

[97]

[118]

[98]

[119]

[99]

[120]

[100]

[121]

[101]

[122]

ARTICLE IN PRESS

290 J.-J. Orteu / Optics and Lasers in Engineering 47 (2009) 282–291 In: Proceedings of 42nd international conference of experimental stress analysis (EAN), Kasperske Hory, Czech Republic, 2004. Nam TH. Mechanical properties of the composite material with elastomeric matrix reinforced by textile cords. PhD thesis, Technical University of Liberec, Czech Republic; 2004. Nelson DV, Makino A, Schmidt T. Residual stress determination using hole drilling and 3D image correlation. Exp Mech 2006;46(1): 31–8. Nicolella DP, Nicholls AE, Lankford J, Davy DT. Machine vision photogrammetry: a technique for measurement of microstructural strain in cortical bone. J Biomechanics 2001;34(1):135–9. Orteu J-J, Garric V, Devy M. Camera calibration for 3D reconstruction: application to the measure of 3D deformations on sheet metal parts. In: SPIE proceedings of European symposium on lasers, optics and vision in manufacturing (EUROPTO), vol. 3101, Munich, Germany, 1997. Orteu J-J, Cutard T, Garcia D, Cailleux E, Robert L. Application of stereovision to the mechanical characterisation of ceramic refractories reinforced with metallic bres. Strain 2007;43(2):96–108. Rechenmacher AL, Medina-Cetina Z. Digital-imaging based measurement of deformed shapes of axisymmetric soil specimens. In: Proceedings of 16th ASCE engineering mechanics conference, Seattle, WA, USA, 2003. Rechenmacher AL. Imaging-based experimental soil mechanics. In: Yamamuro JA, Koseki J, editors. Geomechanics: testing, modeling, and simulation. American Society of Civil Engineers (ASCE); 2005. Revilock DM, Thesken JC, Schmidt TE. 3D digital image correlation of a composite overwrapped pressure vessel during hydrostatic pressure tests. In: Proceedings of SEM 2007 annual conference on experimental and applied mechanics, Springeld, MA, USA, 2007. Richards RG, Wieland M, Textor M. Advantages of stereo imaging of metallic surfaces with low voltage backscattered electrons in a eld emission scanning electron microscope. J Microsc 2000;199(2):115–23. Robert L, Nazaret F, Cutard T, Orteu J-J. Use of 3-D digital image correlation to characterize the mechanical behavior of a ber reinforced refractory castable. Exp Mech 2007;47(6):761–73. Rotrou Y, Sentenac T, Robert L, Orteu J-J. An innovative method for 3D shape/displacement/strain and temperature measurement using a single sensor. In: Proceedings of photomechanics 2006, Clermont-Ferrand, France, 2006. Schmidt TE, Tyson J, Galanulis K. Full-eld dynamic displacement and strain measurement using advanced 3D image correlation photogrammetry: part I. Exp Tech 2003;27(3):47–50. Schmidt TE, Tyson J, Galanulis K. Full-eld dynamic displacement and strain measurement—specic examples using advanced 3D image correlation photogrammetry: part II. Exp Tech 2003;27(4): 22–6. Schmidt TE, Tyson J, Revilock DM, Padula S, Pereira JM, Melis M, et al. Performance verication of 3D image correlation using digital high-speed cameras. In: Proceedings of SEM 2005 annual conference on experimental and applied mechanics, Portland, OR, USA, 2005. Schreier HW. Investigation of two and three-dimensional image correlation methods with applications in mechanical engineering. PhD thesis, University of South Carolina, Columbia, USA; September 2003. Schreier HW, Garcia D, Sutton MA. Advances in light microscope stereo vision. Exp Mech 2004;44(3):278–88. Siebert T, Becker T, Splitthof K. High speed digital image correlation techniques. In: Proceedings of society of automotive engineers (SAE) world congress—SAE technical paper in SP-2033, Detroit, MI, USA, 2006. Siebert T, Becker T, Spiltthof K, Neumann I, Krupka R. Highspeed digital image correlation: error estimations and applications. Opt Eng 2007;46(5):051004. (7pp). [141] Splitthof K, Stecklum S. 3D digital image correlation. VDI-Berichte 2004;1844:209–16. [142] Stanford B, Albertani R, Ifju P. Static nite element validation of a exible micro air vehicle. Exp Mech 2007;47(2):283–94. [143] Sutton MA, Turner JL, Chao YJ, Bruck HA, Chae TL. Experimental investigations of three-dimensional effects near a crack tip using computer vision. Int J Frac 1992;53(3):201–28. [144] Sutton MA, McNeill SR, Helm JD, Boone ML. Measurement of crack tip opening displacement and full-eld deformations during fracture of aerospace materials using 2D and 3D image correlation methods. In: Lagarde A, editor. IUTAM symposium on advanced optical methods and applications in solid mechanics. Solid mechanics and its applications, vol. 82. Netherlands: Springer; 1998. p. 571–80. [145] Sutton MA, McNeill SR, Helm JD, Chao YJ. Advances in twodimensional and three-dimensional computer vision. In: Rastogi PK, editor. Photomechanics, Topics in applied physics. Berlin: Springer; 2000. [146] Sutton MA, Yan J, Deng X, Cheng C-S, Zavattieri P. Threedimensional digital image correlation to quantify deformation and crack-opening displacement in ductile aluminum under mixed-mode I/III loading. Opt Eng 2007;46(5):051003. (17pp). [147] Synnergren P. Measurement of three-dimensional displacement elds and shape using electronic speckle photography. Opt Eng 1997;36(8):2302–10. [148] Synnergren P. Stereovision setup for accurate 3D deformation measurements. In: Yamaguchi I, editor. SPIE proceedings of optical engineering for sensing and nanotechnology, vol. 3740, Yokohama, Japan, 1999. p. 74–7. [149] Synnergren P, Sjodahl M. A stereoscopic digital speckle photo¨ graphy system for 3-D displacement eld measurements. Opt Lasers Eng 1999;31(6):425–43. [150] Synnergren P, Goldrein HT. Dynamic measurements of internal three-dimensional displacement elds with digital speckle photography and ash X rays. Appl Opt 1999;38(28):5956–61. [151] Synnergren P. White light and X-ray digital speckle photo graphy. PhD thesis, Lulea University of Technology, Sweden; 2000. [152] Synnergren P, Berglund T, Soderkvist I. Estimation of curl in paper ¨ using a combination of shape measurement and least-squares modelling. Opt Lasers Eng 2001;35(2):105–20. [153] Tiwari V, Sutton MA, McNeill SR. Assessment of high speed imaging systems for 2D and 3D deformation measurements: methodology development and validation. Exp Mech 2007;47(4): 561–79. [154] Tong W. Plastic surface strain mapping of bent sheets by image correlation. Exp Mech 2004;44(5):502–11. [155] Tyson J, Schmidt T, Galanulis K. Advanced photogrammetry for robust deformation and strain measurement, In: Proceedings of SEM 2002 annual conference on experimental and applied mechanics, Milwaukee, WI, USA, 2002. [156] Tyson J, Schmidt T, Galanulis K. Biomechanics deformation and strain measurements with 3D image correlation photogrammetry. Exp Tech 2002;26(5):39–42. [157] Vialettes P, Siguier J-M, Guigue P, Karama M, Mistou S, Dalverny O, et al. Visco-elastic laws study for mechanical modelization of super-pressure balloons. In: Proceedings of AIAA fth aviation, technology, integration, and operations (ATIO) and 16th lighterthan-air systems technology and balloon systems conferences, Arlington, VA, USA, 2005. [158] Vialettes P, Siguier J-M, Guigue P, Karama M, Mistou S, Dalverny O, et al. Experimental and numerical simulation of super-pressure balloon apex section: mechanical behavior in realistic ight conditions. Adv Space Res 2006;37(11):2082–6. [159] Vignon F, Le Besnerais G, Boivin D, Pouchou J-L, Quan L. 3D reconstruction from scanning electron microscopy using stereovision and self-calibration. In: Proceedings of physics in signal and image processing, Marseille, France, 2001.

[123]

[124]

[125]

[126]

[127]

[128]

[129]

[130]

[131]

[132]

[133]

[134]

[135]

[136]

[137]

[138] [139]

[140]

ARTICLE IN PRESS

J.-J. Orteu / Optics and Lasers in Engineering 47 (2009) 282–291 [160] Vogel JH, Lee D. An automated two-view method for determining strain distributions on deformed surfaces. J Mater Shaping Technol 1989;6(4):205–16. [161] Walz T, Ettemeyer A. Material and component validation by speckle interferometry and correlation methods. Insight NonDestructive Testing Condition Monitoring 2005;47(4):226–31. [162] Wasil A, Merchant DC. Plate-deection measurement by photogrammetric methods. Exp Mech 1964;4(3):77–83. 291

Commercial software

[163] Vic-3D commercial software. Correlated Solutions Incorporated. hhttp://www.correlatedsolutions.com/i. [164] ARAMIS commercial software. GOM, hhttp://www.gom.com/i. [165] StrainMaster commercial software. LaVision, hhttp://www.lavision.de/i. [166] Q-400 commercial software. Dantec Dynamics. hhttp://www. dantecdynamics.com/i.

赞助商链接

相关文章:

- Proposal
- [2] BELHUMEUR P,KRIEGMAN
*D*,YUILLE A. The bas-relief ambiguity [J] . International Journal of*Computer**Vision*,1999,35 ( 1 ) :33-44. [*3*] BAKSHI...

- Web-Scale Computer Vision using MapReduce for Multi...
*Experimental*results for the k-means clustering and singleGaussian background ...[Image Processing and*Computer**Vision*]: General;*D*.1.*3*[Programming ...

- Open Source Computer Vision Library
- 搜索文档 搜索论坛 Open Source
*Computer**Vision*Library VC 2008 Express 下...将在*D*:\Program Files\OpenCV2.*3*.1\build\my 目录下生成 OpenCV.sln 的 ...

- Lesson 21 Computer Vision
- Lesson 21
*Computer**Vision**Computer**Vision*means artificial sight by means ...]有希望的, 有前途的*3*acquisition[??kwi?zi??n] 获得, 获得物 contrast...

- Introduction To Computer Vision Using OpenCV_图文
- Introduction To
*Computer**Vision*Using OpenCV_信息与...Haar-like features*in*a 2-*D*image include ...*3*.Only one response to a single edge: This ...

- 顶级期刊排名
*Computer**Vision*IEEE Trans on Neural Networks ...*Experimental*and Theoretical Artificial Intelligence ...*3*0 计算机视觉、IP、AI 方向的一些顶级会议和期刊...

- 关于计算机视觉(ComputerVision, CV)相关领域的网站
- 以下链接是本人整理的关于计算机视觉(
*ComputerVision*, CV)相关领域的网站链接,其中 有 CV 牛人的主页,CV 研究小组的主页,CV 领域的 paper,代码,CV 领域的最新动态...

- 计算机视觉领域牛科研团队(machine vision group)
- //
*vision*.stanford.edu/research.html*3*).加州大学...San Diego.*computer**vision*--http://*vision*.ucsd...人物 Tomasi,Kana*d*--http://www.cs.cmu.edu/~...

- CCD stereo vision measurement system theory
*computer**vision*technology is an important branch of*computer**vision*has been...= Z - ^ Z (9)*3*System parameters analysis and*experimental*results*3*....

- Scrap metal sorting with colour vision and inductiv...
- The
*computer**vision*setup is based on the colour...6.*Experimental*Results Two sorting machines (...[*3*] M. S. Stachowicz and*D*. Lemke, Image ...

更多相关标签:

- Kernel Methods in Computer Vision
- Markov Models in Computer Vision
- Advances in Computer Vision(ppt)
- 3. Single Camera Models and Calibration Procedures in Computer Vision
- Computer Vision Face Tracking For Use in a Perceptual User Interface
- Particle Filter and Applications in Computer Vision, Tracking
- MATHEMATICAL MODELS FOR IMAGE SEGMENTATION IN COMPUTER VISION
- 3-D Computer Vision Using Structured Light
- Control of polygonal mesh resolution for 3-D computer vision
- EFFICIENT GRAPH-BASED ENERGY MINIMIZATION METHODS IN COMPUTER VISION
- Experimental status of quaternionic quantum mechanics
- Recovering estimate of fluid flow from image sequence data. Computer vision and Image under
- Vision-based interpretation of hand gestures for remote control of a computer mouse
- experimental-mechanics-1
- Experimental Mechanics DOI 10.1007s11340-007-9046-8 Flexible Probes for Characterizing Surf